Please use this identifier to cite or link to this item:
|Title:||Attenuation of a Non-parallel Beam of Gamma Radiation by Thick Shielding- application to the Determination of the 235U Enrichment with NaI Detectors|
|Authors:||MORTREAU PATRICIA; BERNDT REINHARD|
|Citation:||NUCLEAR INSTRUMENTS & METHODS IN PHYSICS RESEARCH SECTION A-ACCELERATORS SP vol. 550 p. 675-690|
|Publisher:||ELSEVIER SCIENCE BV|
|Type:||Articles in periodicals and books|
|Abstract:||The traditional method used to determine the Uranium enrichment by nondestructive analysis is based on the “enrichment meter principle” . It involves measuring the intensity of the 186 keV net peak area of 235U in “quasi-infinite” samples. A prominent factor, which affects the peak intensity, is the presence of gamma absorbing material (e.g., container wall, detector cover) between the sample and the detector. Its effect is taken into consideration in a commonly called “wall thickness” correction factor. Often calculated on the basis of approximations, its performance is adequate for small attenuation factors applicable to the case of narrow beams. However these approximations do not lead to precise results when wide non-parallel beams are attenuated through thick container walls. This paper is dedicated to the calculation by numerical integration of the geometrical correction factor (Kwtc) which describes the effective mean path length of the radiation through the absorbing layer. This factor was calculated as a function of various measurement parameters (types and dimensions of the detector, of the collimator and of the shielding) for the most commonly used collimator shapes and detectors. Both coherent scattering (Rayleigh) and incoherent scattering (Compton) are taken into account for the calculation of the radiation interaction within the detector.|
|JRC Institute:||Space, Security and Migration|
Files in This Item:
There are no files associated with this item.
Items in repository are protected by copyright, with all rights reserved, unless otherwise indicated.