Please use this identifier to cite or link to this item:
|Title:||Development and Validation of an Analytical Methos for the Determination of Semicarbazide in fresh Egg and in Egg Powder based on the use of LC-MS/MS|
|Authors:||SZILAGYI SZILARD; DE LA CALLE GUNTINAS MARIA BEATRIZ|
|Citation:||ANALYTICA CHIMICA ACTA vol. 572 no. 1 p. 113-120|
|Type:||Articles in periodicals and books|
|Abstract:||In this work the development of a method to determine semicarbazide (SEM) in fresh whole egg and in whole egg and egg white powders as those used in the food industry, by using LC-MS/MS is described. The method is based on a recently validated one for the determination of SEM in baby food. SEM is extracted from egg with 0.2 mol.L-1 hydrochloric acid and derivatised with 2 nitrobenzaldehyde. The extract is neutralised and purified by passing through a solid phase extraction (SPE) cartridge. The 2-nitrobenzaldehyde derivative is eluted with ethyl acetate. The eluate is evaporated to dryness and the residue re-dissolved in methanol: water mobile phase. SEM is determined by reversed phase LC-MS/MS. The “in-house” validation of this method has been performed taking into account the “Harmonized guidelines for single-laboratory validation of methods of analysis” (IUPAC Technical report) and the Commission Decision (2002/657/EC). The performance characteristics of the method were established by in-house validation procedures employing assays with standard solutions, sample blanks and spiked samples. Linearity, matrix effect, trueness, precision, selectivity, limits of detection and of quantification were determined. The fitness for purpose of this method was assessed based on its performance characteristics. LOD’s of 0.15 µg•kg-1 and 0.4 µg•kg-1 for whole fresh egg and industrial egg powders, respectively, were obtained. The LOQ for fresh whole egg was 0.2 µg•kg-1 and for industrial egg powders 0.8 µg•kg-1. Linear calibration curves were obtained in the ranges 1-100 µg•kg-1 and 0.8-400 µg•kg-1 for fresh whole egg and the egg powders, respectively.|
|JRC Institute:||Institute for Reference Materials and Measurements|
Files in This Item:
There are no files associated with this item.
Items in repository are protected by copyright, with all rights reserved, unless otherwise indicated.