Please use this identifier to cite or link to this item:
|Title:||Determination of Sucralose in Soft Drinks by High-Performance Thin-Layer Chromatography: Interlaboratory Study|
|Authors:||STROKA Joerg; DONCHEVA Ivanka; SPANGENBERG Bernd|
|Citation:||JOURNAL OF AOAC INTERNATIONAL vol. 92 no. 4 p. 1153-1159|
|Type:||Articles in periodicals and books|
|Abstract:||An interlaboratory comparison was carried out to evaluate the effectiveness of a method based on HPTLC in which reagent-free derivatization is followed by UV/fluorescence detection. The method was tested for the determination of sucralose (C12H19Cl3O8; (2R,3R,4R,5S,6R)-2-[(2R,3S,4S,5S)-2,5-bis(chloromethyl)-3,4-dihydroxyoxolan-2-yl]oxy-5-chloro-6-hydroxymethyl)oxane-3, 4-diol; CAS Registry No. 56038-13-2) in carbonated and still beverages at the proposed European regulatory limits. For still beverages, a portion of the sample was diluted with methanol¿water. For carbonated beverages, a portion of the sample was degassed in an ultrasonic bath before dilution. Turbid beverages were filtered after dilution through an HPLC syringe filter. The separation of sucralose was performed by direct application on amino-bonded (NH2) silica gel HPTLC plates (no cleanup needed) with the mobile phase acetonitrile¿water. Sucralose was determined after reagent-free derivatization at 190°C; it was quantified by measurements of both UV absorption and fluorescence. The samples, both spiked and containing sucralose, were sent to 14 laboratories in five different countries. Test portions of a sample found to contain no sucralose were spiked at levels of 30.5, 100.7, and 299 mg/L. Recoveries ranged from 104.3 to 124.6% and averaged 112% for determination by UV detection; recoveries ranged from 98.4 to 101.3% and averaged 99.9% for determination by fluorescence detection. On the basis of the results for spiked samples (blind duplicates at three levels), as well as sucralosecontaining samples (blind duplicates at three levels and one split level), the values for the RSDr ranged from 10.3 to 31.4% for determinations by UV detection and from 8.9 to 15.9% for determinations by fluorescence detection. The values for the RSDR values ranged from 13.5 to 31.4% for determinations by UV detection and from 8.9 to 20.7% for determinations by fluorescence detection.|
|JRC Directorate:||Health, Consumers and Reference Materials|
Files in This Item:
There are no files associated with this item.
Items in repository are protected by copyright, with all rights reserved, unless otherwise indicated.