Please use this identifier to cite or link to this item:
|Title:||Extension of the Calibration Curve for the PGRA Facility in Petten|
|Authors:||NIEVAART Victor Alexander; STECHER-RASMUSSEN Finn; SAUERWEIN Wolfgang; WITTIG A.; MOSS Raymond; APPELMAN K.|
|Citation:||APPLIED RADIATION AND ISOTOPES vol. 67 no. 7-8S p. S362-S364|
|Publisher:||PERGAMON-ELSEVIER SCIENCE LTD|
|Type:||Articles in periodicals and books|
|Abstract:||At the boron neutron capture therapy (BNCT) facility in Petten, the Netherlands, 10B concentrations in biological materials are measured with the prompt gamma ray analyses facility that is calibrated using certified 10B solutions ranging from 0 to 210 ppm. For this study, newly certified 10B solutions ranging up to 1972ppm are added. MCNP simulations of the setup range to 5000 ppm. A second order polynomial (as already used) will fit 10B-concentrations less than 300 ppm. Above 300ppm a fitted third order polynomial is needed to describe the calibration curve accurately.|
|JRC Institute:||Energy, Transport and Climate|
Files in This Item:
There are no files associated with this item.
Items in repository are protected by copyright, with all rights reserved, unless otherwise indicated.