Please use this identifier to cite or link to this item:
|Title:||International Exercise on 124Sb Activity Measurements|
|Authors:||CHAUVENET B.; BE Marie-Martine; AMIOT M.-N.; BOBIN C.; LEPY M.-C.; BRANGER T.; LANIECE I.; LUCA A.; SAHAGIA M.; WAETJEN A. C.; KOSSERT K.; OTT O.; NAEHLE O.; DRYAK P.; SOCHOROVA J.; KOVAR P.; AUERBACH P.; ALTZITZOGLOU Timotheos; POMME Stefaan; SIBBENS Goedele; VAN AMMEL Raf; PAEPEN Jan; IWAHARA A; DELGADO J.u.; POLEDNA R.; DA SILVA C. J.; JOHANSSON L.; STROAK A.; BAILAT C.; NEDJADI Y.; SPRING P.|
|Citation:||APPLIED RADIATION AND ISOTOPES vol. 68 no. 7-8 p. 1207-1210|
|Publisher:||PERGAMON-ELSEVIER SCIENCE LTD|
|Type:||Contributions to Conferences|
|Abstract:||An international exercise, registered as EUROMET project n° 907, was launched to measure both the activity of a solution of 124Sb and the photon emission intensities of its decay. The same solution was sent by LNE-LNHB to eight participating laboratories. This paper presents the results of the first part of the exercise dealing with the comparison of activity measurements. In order to identify possible biases, the participants were asked to use all possible activity measurement methods available in their laboratory and then to determine their reference value for comparison. Thus measurement results were given using 4 pi beta-gamma coincidence/anti-coincidence counting, CIEMAT/NIST liquid scintillation counting, 4 pi-gamma counting with well-type ionization chambers and well-type crystal detectors. The results are compared and show a maximum discrepancy of about 1.6%, possible explanations are sought.|
|JRC Institute:||Institute for Reference Materials and Measurements|
Files in This Item:
There are no files associated with this item.
Items in repository are protected by copyright, with all rights reserved, unless otherwise indicated.