Please use this identifier to cite or link to this item:
|Title:||Determination of 240Pu/239Pu, 241Pu/239Pu, 242Pu/239Pu Isotope Ratios in Environmental Reference Materials and Samples from Chernobyl by Thermal Ionization Mass Spectrometry (TIMS) and Filament Carburization|
|Authors:||JAKOPIC Rozle; RICHTER Stephan; KUEHN Heinz; AREGBE Yetunde|
|Citation:||JOURNAL OF ANALYTICAL ATOMIC SPECTROMETRY vol. 25 no. 6 p. 815-821|
|Publisher:||ROYAL SOC CHEMISTRY|
|Type:||Articles in periodicals and books|
|Abstract:||In various areas such as e.g. nuclear safeguards, environmental monitoring, nuclear forensics, precise, sensitive and accurate plutonium analysis are needed. Plutonium isotope ratios are used as "fingerprints" in revealing different sources of plutonium contamination in the environment. In this study, 240Pu/239Pu, 241Pu/239Pu and 242Pu/239Pu isotope ratios were determined in four different reference materials and samples from Chernobyl using Thermal Ionization Mass Spectrometry (TIMS) in combination with multiple ion counting (MIC) and filament carburization. After the samples were leached with 8M HNO3, plutonium was co-precipitated on CaC2O4 and separated from interfering radionuclides and matrix elements by anion exchange and extraction chromatography (TEVA Spec). The purified plutonium solution was loaded on carburized Re filaments for TIMS measurements. Filament carburization using benzene gas as a pure carbon source was applied in order to enhance the efficiency of plutonium for the mass spectrometric analysis. The carburized Re filaments were prepared in a special vacuum chamber in which benzene was introduced at a pressure of 5¿10-3 mbar and a current of 4A for 1 hour. The TIMS measurements were performed using a multiple ion counting (MIC) detection system using both multi-dynamic and static measurement modes. The analytical procedure was validated using various reference materials and further applied to soil and moss samples from Chernobyl. The obtained 240Pu/239Pu isotope ratios for reference materials were in good agreement with the literature values. Due to the newly developed techniques and procedures, data not only for the240Pu/239Pu "major" isotope ratios but also for the first time 241Pu/239Pu and 242Pu/239Pu "minor" isotope ratios were acquired with lower uncertainties, which is particularly important for fingerprinting of environmental samples in nuclear safeguards.|
|JRC Institute:||Institute for Reference Materials and Measurements|
Files in This Item:
There are no files associated with this item.
Items in repository are protected by copyright, with all rights reserved, unless otherwise indicated.