Please use this identifier to cite or link to this item:
|Title:||Determination of Dissolved Bromate in Drinking Water by Ion Chromatography and Post Column Reaction: Interlaboratory Study|
|Authors:||CORDEIRO RAPOSO Fernando; ROBOUCH Piotr; DE LA CALLE GUNTINAS Maria Beatriz; SCHMITZ Franz|
|Citation:||JOURNAL OF AOAC INTERNATIONAL vol. 94 no. 5 p. 1592-1600|
|JRC Publication N°:||JRC61235|
|Type:||Articles in Journals|
|Abstract:||A collaborative study, International Evaluation Measurement Programme®-25a, was conducted in accordance with international protocols to determine the performance characteristics of an analytical method for the determination of dissolved bromate in drinking water. The method should fulfill the analytical requirements of Council Directive 98/83/EC (referred to in this work as the Drinking Water Directive; DWD). The new draft standard method under investigation is based on ion chromatography followed by post-column reaction and UV detection. The collaborating laboratories used the Draft International Organization for Standardization (ISO)/Draft International Standard (DIS) 11206 document. The existing standard method (ISO 15061:2001) is based on ion chromatography using suppressed conductivity detection, in which a preconcentration step may be required for the determination of bromate concentrations as low as 3 to 5 μg/L. The new method includes a dilution step that reduces the matrix effects, thus allowing the determination of bromate concentrations down to 0.5 μg/L. Furthermore, the method aims to minimize any potential interference of chlorite ions. The collaborative study investigated different types of drinking water, such as soft, hard, and mineral water. Other types of water, such as raw water (untreated), swimming pool water, a blank (named river water), and a bromate standard solution, were included as test samples. All test matrixes except the swimming pool water were spiked with high-purity potassium bromate to obtain bromate concentrations ranging from 1.67 to 10.0 μg/L. Swimming pool water was not spiked, as this water was incurred with bromate. Test samples were dispatched to 17 laboratories from nine different countries. Sixteen participants reported results. The repeatability RSD (RSDr) ranged from 1.2 to 4.1%, while the reproducibility RSD (RSDR) ranged from 2.3 to 5.9%. These precision characteristics compare favorably with those of ISO 15601. A thorough comparison of the performance characteristics is presented in this report. All method performance characteristics obtained in the frame of this ollaborative study indicate that the draft ISO/DIS 11206 standard method meets the requirements set down by the DWD. It can, therefore, be considered to fit its intended analytical purpose.|
|JRC Institute:||Institute for Reference Materials and Measurements|
Files in This Item:
There are no files associated with this item.
Items in repository are protected by copyright, with all rights reserved, unless otherwise indicated.