Please use this identifier to cite or link to this item:
|Title:||Assessment of the legislated particle number measurement procedure for a Euro 5 and a Euro 6 compliant diesel passenger cars under regulated and unregulated conditions|
|Authors:||MAMAKOS ATHANASIOS; MARTINI Giorgio; MANFREDI Urbano|
|Citation:||JOURNAL OF AEROSOL SCIENCE vol. 55 p. 31-47|
|Publisher:||ELSEVIER SCI LTD|
|Type:||Articles in periodicals and books|
|Abstract:||We investigated the possibility of extending the legislated particle number (PN) Measurement procedure to (a) control diesel exhaust emissions under unregulated Driving conditions, including tests at -7°C and during regeneration of the particulate filter, and (b) reduce the lowest detectable size below 23 nm, through dedicated tests of a Euro5b and a Euro6 diesel vehicles. The PN emissions of the two vehicles were at or below the limit of 6x1011 #/km under all non-regenerative conditions examined. Active regeneration of the Diesel Particulate Filters (DPFs) occurred more frequently (220–510km) than previously reported, and resulted in elevated PM (upto11mg/km) and PN(1–4x1012 #/km) emissions. Condensation Particle Counters (CPCs) having a nominal 50% detection efficiency (10nm and 4.5nm) below the legislated 23nm, detected 50–60% more non-volatile particles over the certification test cycle (New European Driving Cycle). Even higher excess fractions of nano-sized non-volatile particles (160–500%) were observed under urban driving of the Euro5b vehicle at -7°C, with the absolute levels still remaining below 6x1011 #/km however. DPF regeneration and motor way driving favoured the homogeneous nucleation of volatile particles in the Constant Volume Sampler(CVS), which also contributed to the signals of low cut-off size CPCs sampling downstream of the Volatile Particle Remover (VPR). No evidence of such volatile interference could be observed in the legislated PN procedure. A potential shift of the lowest detectable size towards smaller sizes, will require further developments in the measurement procedure.|
|JRC Institute:||Energy, Transport and Climate|
Files in This Item:
There are no files associated with this item.
Items in repository are protected by copyright, with all rights reserved, unless otherwise indicated.